
->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
The following is the header information for this acquisition
Sample ID                      12
Sample Info                    lys, h2O
Operator                       
Comments                       


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
Log file for data file : MAY 18 2018 17-01
Software : Version 262144.0.000 Feb  8 2013

->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
  #######   EI Ionizer Parameters  ######   
 Electron Energy                            -1
 Electron Emission                          -1 milliamps
 EI Filament (1 is on)                      -1
 ########################################   


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
  ####### Ion Optic Parameters ##########   
 Ion Region                               -1
 Extractor                                -1
 Lens 1 / 3                               -1
 Lens 2                                     -1
 Quad Entrance                              -1
 Quad Exit                                  -1
 ########################################   


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
  #######   Electron Multiplier Parameters  ##################### 
 Multiplier  Volts                                          2500
 Multiplier polarity (0 = negative voltage, 1 = positive voltage) 0
 Multiplier on/off   (0 = off, 1 = on)                            1
 ################################################################ 


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
  #######   Conversion Dynode parameters  ####################### 
 Dynode Volts                                                     5000
 Dynode on/off       (0 = off, 1 = on)                            1
 Dynode polarity     (0 = negative voltage, 1 = positive voltage) 0
 ################################################################ 


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
  ########   Preamp Parameters  ########## 
 The counting preamp was used 
 ########################################## 

->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
 adc_samples / mass actual     8
 adc_Samples / mass requested  100
 Prescan                       30
 Scan time requested           0.2
 Scan time actual              0.63932
 Quad 1 pole reverse           0
 Quad 1 delta res              0


->MAY 18 2018   05:01:07 PM (rt: 0.00 min.)
System:Merlin Automation Data System     Login user:Mass Spec
 The following Ion file descriptor lists positive and negative tables
 The positive tables parameters are appropriate for Q0, Q1, Q2, and Q3 readings
 The negative tables are for a different model Quadrupole Power Supply
 Your ion polarity is determined by the Detector Parameters


********** Scan descriptor **********
Scan descriptor name : 2017-02-03 (modified)
Scan mode    : Profile
SIM width    : 0.700 amu total
Scan time    : 0.200 Sec.
CID Gas      : Off
Ion polarity : Positive ions
  Q1FM    Q1LM    Q3FM    Q3LM    Collision Energy
  10.0  1000.0                  

********** Tune Tables **********
MAY 18 2018   05:01:07 PM

System name: Merlin Automation Data System
Tune table name: init

Positive tables:
Lens 1               Flat      0.00
Lens 2               Flat    260.00
Lens 3               Flat      0.00
Q0 Offset            Flat      0.00
Lens 4               Flat   -140.00
Q1 Offset    (Pass)  Flat      0.00
Q1 Offset   (Filter) Flat    100.00
Q1 Resolution        Flat      0.00
Q1 Calibration       Flat     -1.40
Q2 Offset (Q1MS)     Flat      0.00
Q2 Offset (Q3MS)     Flat    -18.00
Collision Energy     Flat      0.00
Q3 Offset    (Pass)  Flat      0.00
Q3 Offset   (Filter) Flat      0.00
Q3 Resolution        Flat      0.00
Q3 Calibration       Flat      0.00

Negative tables:
Lens 1               Flat     30.00
Lens 2               Flat    150.00
Lens 3               Flat     37.69
Q0 Offset            Flat     30.00
Lens 4               Flat    -11.00
Q1 Offset    (Pass)  Flat      7.00
Q1 Offset   (Filter) Flat      2.62
Q1 Resolution        Flat     -0.25
Q1 Calibration       Flat      0.70
Q2 Offset (Q1MS)     Flat     18.00
Q2 Offset (Q3MS)     Flat     18.00
Collision Energy     Flat      0.00
Q3 Offset    (Pass)  Flat      9.00
Q3 Offset   (Filter) Flat      1.26
Q3 Resolution        Flat     -1.55
Q3 Calibration       Flat      0.80

Started new scan at 0.0 minutes,Positive ions
  Q1FM    Q1LM    Q3FM    Q3LM    Collision Energy
  10.0  1000.0                  

->MAY 18 2018   05:01:25 PM 
Data file header modified! -> Sample Info changed
From:lys, h2O
To  :gly

->MAY 18 2018   07:32:41 PM (rt: 151.57 min.)
System:Merlin Automation Data System     Login user:Mass Spec
Collection stopped
Ctrl macros:                            times called
   LOG_ALL                                         1
   FISH                                        14247
   DO_PLOT_THINGS                              14247
Proc macros:                            times called
   CHRO_CLICK                                     34

